Colorimeter, order code COL-BTA, $115
Specifications and User Guide
-Primary Test: Connect the Colorimeter to a Vernier Interface. Allow the Colorimeter to warm up for 5 minutes. Place a cuvette, about 3/4 full of distilled water, in the Colorimeter and close the lid. Select a wavelength. Press and hold the CAL button on the Colorimeter. When the red LED begins to flash, release the CAL button. When the red LED stops flashing, the Colorimeter is calibrated. Check the absorbance readings, which should be very nearly zero.
-Secondary test: Prepare a solution with a color (a simple test solution for the 635 nm wavelength setting can be made by adding 2 drops of green food coloring to 100 mL of tap water). After calibrating the Colorimeter, fill a cuvette about 3/4 full with the test solution. Place the cuvette in the Colorimeter and close the lid. The absorbance readings should change to a reading between 0.8 - 1.5. You may dilute the test solution and place a cuvette of the dilute solution in the Colorimeter to check the change in absorbance.
-The normal, most useful absorbance range is 0.05 - 1.0, or 10% - 90% transmittance.
Please keep in mind the following when using your colorimeter:
1) When you press the CAL button, make sure the red LED lights flash several times. You may still have to wait a few seconds after the red LED stops flashing before the absorbance reading is ~0.000 or the % transmittance is ~100.0%.
2) If you calibrated with a cuvette of water and then remove this cuvette of water, it is expected that the absorbance value will no longer be 0.000 or the % transmittance may be much lower than 100.0% (even as low as 70%). If you put the cuvette of water back in the Colorimeter, the absorbance will return to ~0.000 or the % transmittance will return to ~100.0%.
3) If you switch to a new wavelength, you need to calibrate the Colorimeter again.
4) If your solution is very dark and you are getting absorbance readings greater than 2.00, then dilute your solution. Solutions should typically be fairly translucent to get accurate readings with the Colorimeter.
RELATED TIL ENTRIES
-TIL 356: How does a colorimeter compare to a spectometer?
-TIL 1921: Do you sell replacement lids for my plastic cuvettes?
-TIL 2539: Why don't the absorbance readings for the Colorimeter or the spectrometers have units?
-TIL 3403: Do I have an old or current Colorimeter?
-TIL 1665: How do I calibrate my older Colorimeter?
-TIL 2915: How do I calibrate a COL-DIN Colorimeter using a stand alone LabQuest 2?
-TIL 784: My Colorimeter is not working properly. What should I do?
-TIL 2297: Can my Colorimeter report in percent Transmittance using a LabQuest?
-TIL 2569: Why don't all of my spectrometers and/or Colorimeters read the same absorbance value for the same sample?
-TIL 2949: Do you have any experiments written for the LabQuest and the older model Colorimeter?
-TIL 3148: How do I run the AP Chemistry food dye lab with a Colorimeter instead of a Spectrometer?
MAINTENANCE and STORAGE
-TIL 3039: Liquid has spilled inside my Colorimeter. What should I do?
-order code: COL-BTA
-Range: 0 to 3 (absorbance)
-Useful Range 0.05 to 1.0 (absorbance) and 90% to 10% transmittance (%T)
-Wavelengths: 430 nm, 470 nm, 565 nm, 635 nm
-Supply voltage: 5VDC +/- 25 mV
-Supply current (typical): 40 mA
-Power up time: 700 ms (maximum)
-Output voltage range: 0 - 4 V
-Transfer function: Vout = 0.035 x (%T) + 0
-Stored calibration values: Slope = 28.571; Intercept = 0
-Calibration? Yes, each time before use. Connect the Colorimeter to an interface or LabQuest 2 (or LabQuest) and run Logger Pro 3 or the LabQuest App. Allow the Colorimeter to warm up for 5 minutes. Select a wavelength. Open the lid, insert a cuvette filled about 3/4 full of distilled water, and close the lid. Make sure that one clear side of the cuvette is lined up with the arrow at the top of the cuvette slot. Press the CAL button and hold it until the red LED begins to flash and then release the CAL button. When the LED stops flashing, the calibration is complete.
RELATED VERNIER PRODUCTS
- for extra cuvettes, Plastic Cuvettes (Visible Range), order code CUV, $19 (pkg. of 100)