Primary Test: Plug the probe into an interface and run the data collection program. Use wire leads to connect the probe to a DC supply with a known resistance. Use a voltage probe or voltmeter to measure the voltage. Compare the measured current against the theoretical reading. We recommend a DC battery for this test, since some DC power supplies may not deliver clean DC voltage.
-new order code: DCP-BTA
-old order code: none (use CV-DIN)
-Current Probe Range: +/-0.6A
-Maximum Voltage on any input: +/-10 V
-Input Impedance (between inputs): 0.1Ω
-Input Impedance (to ground): 10 MΩ
-Resolution, 10-bit (CBL, CBL 2, original ULI): 1.25 mA
-Resolution, 12-bit (LabPro, Go! Link, ULI II, Serial Box): 0.31 mA
-Supply Voltage: 5 VDC
-Supply Current (typical): 9 mA
-Output Voltage Range: 0 to 5 V
-Transfer Function: Vout = -4(I) = 2.5
-Calibrate? No. The probe is set to match the stored calibration before shipping.
-If you choose to calibrate the probe, conduct a two-point calibration using two known currents.
-You may also zero the probe by shorting out the leads and then choosing the Zero option in the data collection program. If you zero the probe, it will adjust the calibration offset but does not adjust the calibration gain.