-Primary Test: Connect an ORP sensor or a pH sensor to the Electrode Amp and connect the Electrode Amp to a LabPro. Start the data-collection program. Place the ORP or pH sensor in a buffer solution of known pH and confirm that the readings are within range.
-Secondary Test: Confirm that the BNC connection is secure.
-new order code: EA-BTA
-old order code: none
-Power = 7 mA @ 5VDC
-Input range = –450mV to +1100 mV
-Impedance > 100M Ohm
-Gain: 2.2 V/V
-Offset: 1.20 V (nominal)
-Calibrate? Normally, yes. However, it depends on the sensor that is connected. The Electrode Amplifier amplifies the voltage produced by an electrode into a range where it can be monitored by the lab interface.
The Electrode Amplifier will report values in units of millivolts. If a pH electrode is attached, however, you can choose to read in units of pH. Custom, linear 2-point calibrations can also be performed if desired.
-You can open an experiment file in LoggerPro which will use the stored calibration in the software.