Calcium Ion-Selective Electrode, order code CA-BTA, $199
Specifications and User Guide
-Primary Test: Check the raw voltages while calibrating in the standards. The high (1000 mg/L) should be around 1.9 volts. The low (10 mg/L) should be around 1.5 volts.
-Secondary Test: Refer to TIL 665: General tips for using Ion Selective Electrodes (ISE).
RELATED TIL ENTRIES
-TIL 1060: Do any of your sensors require special storage?
-TIL 759: Do you sell ISE Standards?
-TIL 665: General tips for using Ion Selective Electrodes (ISE).
-TIL 1296: How can I have my ISE read mV output instead of mg/L?
-TIL 1463: What Is the Difference Between Auto-ID Sensors, Non-Auto-ID Sensors, and Smart Sensors?
-TIL 3389: How do I sterilize an Ion-Selective Electrode?
-Order code: CA-BTA
-Range: 1 to 40,000 mg/L or ppm
-pH range: 3 to 10
-Interfering ions: Pb2+, Hg2+, Sr2+, Cu2+, Ni2+
-Electrode slope: +28 mV/decade
-Approximate calibration voltages: High (1000 mg/L) 1.9 V; Low (10 mg/L) 1.5 V
-Accuracy: ± 10% of full scale (calibrated 10 to 1000 mg/L)
-Temperature range (can be placed in): 0 to 40°C (no temperature compensation)
-Minimum immersion: 1.1 inch
-Electrode length: 155 mm
-Body diameter: 12 mm
-Cap diameter: 16 mm
-Cable length: 100 cm
-Calibrate? Yes. Conduct a two-point calibration, using the standard solutions that are shipped with the sensor. First point: The ISE should be soaking in the High Standard. Enter the concentration value of the High Standard. Second point: Remove the ISE from the High Standard, rinse well with distilled water from a wash bottle, and gently blot dry with a paper towel or lab wipe. Place the electrode into the Low Standard. Wait for the voltage to stabilize. Enter the concentration value of the Low Standard.
-Important: Make sure the ISE is not resting on the bottom of the container, and that the small white reference contacts are immersed. Make sure no air bubbles are trapped below the ISE. Hold the ISE still and wait for the voltage reading to stabilize. Stirring the solution during this process is ideal.
RELATED VERNIER PRODUCTS
-ISE Calcium Replacement Module, Calcium Replacement Module, order code CA-MOD, $79
-ISE Calcium Low Standard, Standard Low Ca ISE Solution, order code CA-LST, $18
-ISE Calcium High Standard, Standard High Ca ISE Solution, order code CA-HST, $18