-Primary Test: Place the toggle switch at the desired setting, run the data-collection program, and take a reading in a solution of known TDS. We suggest using the calibration solution that comes with the probe. It is 500 mg/L TDS or 1000 microsiemens/cm.
-Secondary Test: If you change the setting of the toggle switch while the data collection program is running, reset the program so that the calibration equation for the new setting is selected.
RELATED TIL ENTRIES
–Will I get interference when I put two sensors in the same solution?
–How do I calibrate a sensor in Logger Pro?
–How do I store the calibration for my sensor using Logger Pro?
–Where do I get Conductivity Standards?
–What material are your Conductivity Probes made of?
–Does the conductivity probe have different calibrations?
–How is the temperature compensation calculated on my conductivity probe (CON-BTA)?
–What chemicals can I use with your conductivity probes?
TECH TIPS VIDEO
-order code: CON-BTA
-Range of Conductivity Probe:
-Low Range: 0 to 200 microsiemens/cm (0 to 100 mg/L TDS)
-Mid Range: 0 to 2000 microsiemens/cm (0 to 1000 mg/L TDS)
-High Range: 0 to 20,000 microsiemens/cm (0 to 10,000 mg/L TDS)
-Resolution (with LabPro, Go! Link, ULI II, and Serial Box):
-Low Range: 0.1 microsiemen/cm (0.05 mg/L TDS)
-Mid Range: 1 microsiemen/cm (0.5 mg/L TDS)
-High Range 10 microsiemen/cm (5 mg/L TDS)
-Accuracy using factory calibration: +/-8% of full-scale reading for low range
+/-3% of full-scale reading for mid range
+/-4% of full-scale reading for high range
-Accuracy using custom calibration: +/-2% of full-scale reading for low range
-Response Time: 98% of full-scale reading in 5 seconds, 100% of full-scale in 15 seconds
-Temperature Compensation: automatic from 5 to 35 deg C
-Temperature Range (can be placed in): 0 to 80 deg C
-Cell Constant: 1.0 1/cm
-Description: ABS body, parallel carbon (graphite) electrodes
-Dimensions: 12 mm OD amd 150 mm length
-You do not have to perform a new calibration when using the Conductivity Probe for most experiments in the classroom. However, if your experimental application requires more accurate readings, you should calibrate your sensor.
-See the user manual for calibration instructions: Specifications and User Guide
RELATED VERNIER PRODUCTS
-Conductivity standard solution, low range, Conductivity Standard Solution (Low) (CON-LST) (500 mL)
-Conductivity standard solution, mid range, Conductivity Standard Solution (Middle) (CON-MST) (500 mL)
-Conductivity standard solution, high range, Conductivity Standard Solution (High) (CON-HST) (500 mL)