A non-auto-ID sensor must be manually identified, regardless of the data collection program in use.
An auto-ID sensor is automatically identified when the sensor is connected to a Vernier interface.
A Smart Sensor is a special auto-ID sensor. It has internal memory that allows the user to store a custom calibration to the sensor. A custom calibration can be read and used with any data collection program. A custom calibration can only be stored to the sensor memory via Logger Pro 3.3 or newer or a LabQuest.
As new models of each sensor are introduced, a feature that is often added is the ability to Auto ID or to be a Smart Sensor. Many older sensors are not smart.
As of May of 2016, the following new sensors are non-auto-ID:
1) all sensors with a DIN connector
2) Vernier Structure and Materials Tester Displacement Sensor
The following sensors will not AutoID on a LabPro:
1) Drop Counter
2) Radiation Monitor
3) Rotary Motion
4) Digital Control Unit
5) Motion Encoder
6) Polarimeter (Chemical)
7) Diffraction Apparatus
8) Projectile Launcher
(Note some of these may require a firmware update to AutoID on LabQuest.)
The following sensors are auto-ID, but not Smart Sensors:
2) Differential Voltage
4) Exercise Heart Rate
5) Hand Grip Heart Rate
6) Stainless Steel Temperature
7) Surface Temperature
8) Extra-Long Temperature
9) TI Light (packaged with the original CBL), also sold as TILT-BTA
10) Voltage (-10 to +10 V)
11) +-30 Voltage
13) Motion Detector
15) Drop Counter
16) Radiation Monitor
17) Rotary Motion
18) Digital Control Unit
19) Motion Encoder
20) Polarimeter (Chemical)
21) Diffraction Apparatus
22) Projectile Launcher
|Channel 1, 2, 3 and 4|
|IDENT Value1||Sensor Type||Range|
|2.2K||Thermocouple °C||-20°C to 1400 °C|
|33K||TI Voltage sensor||10 to +10 Volts|
|6.8K||Current sensor1||-10 to +10 Amps|
|3.3K||Resistance sensor||1K to 100 K?|
|22K||Extra long temperature||50 °C to 150 °C|
|68K||CO2 gas sensor (PPM)||0 to 5000 ppm|
|100K||Oxygen gas sensor(PCT)||0 to 27%|
|150K||C V voltage sensor (V)||-6 to +6 Volts|
|220K||C V current sensor (A)||-0.6 to +0.6 Amps|
|10K||Stainless steel or TI||-25 °C to 125 °C|
|15K||Stainless steel or TI||-13 °F to 257 °F|
|4.7K||TI Light sensor||0 to 1|
|1K||Ex heart rate sensor||N/A|
|47K||Voltage sensor||0 to 5 Volts|
|DIG 1 and DIG 2 (SONIC)|
|IDENT Value1||Sensor Type||Range|
|15K||Motion detector, meters||0.4 meter to 6 meters|
|22K||Motion detector, meters||0.4 meter to 6 meters|
|10K||Motion detector, feet||1.4 feet to 18 feet|
When you use an auto-ID sensor (whether or not it is also smart) with any of our data collection programs listed above (Logger Pro, LabQuest, DataPro, Datamate, or EasyData), not only is the sensor identified by the program but it uses a factory calibration.
IMPORTANT NOTE: The factory calibration of our sensors is very precise. For most experiments, it is not necessary to calibrate a sensor.
After you calibrate a sensor, you may choose to write down the calibration equation (slope and Y-intercept). Then, if you wish to use that calibration equation for the sensor in a future experiment, you can manually enter the slope and Y-intercept instead of conducting a new calibration.
If you have calibrated a sensor in Logger Pro or on LabQuest, you can also save the calibration equation by saving the experiment file. You can use the saved file later to conduct experiments with the sensor and the calibration equation will be used as well.
With smart sensors, after you calibrate the sensor you can store the calibration equation on the sensor itself. This is what makes the smart sensors °smart°. Remember that this can only be done via computer and Logger Pro 3.3 or newer or LabQuest. Follow the steps below to complete this process.
Logger Pro 3 Instructions
1) Connect the sensor to a computer, via an interface, and start the Logger Pro software.
2) Open “Calibrate â†’ LabPro CH1” under the Experiment menu.
3) Click Calibrate Now.
4) Complete the calibration for the sensor.
5) After calibration, click the Calibration Storage tab along the top of the window.
6) Click Set Sensor Calibration.
7) On the Set Sensor Calibration window, make sure that the Default Page is the same as the Add Page. Normally, both Add page and Default Page will be Page 1. Click Set.
8) On the Sensor Settings window, click Done.
9) On the Writing Sensor Memory window, click Write.
Please note that you can always set the sensor back to its factory settings by opening the °Live Calibration° window, selecting °Calibration Storage° from the drop down menu, and clicking °Write Factory Default to Sensor°.
For more detailed documentation on storing custom calibrations on smart sensors, including screen shots, see Logger Pro‘s built-in help file.