This information pertains to the original LabQuest.
If you have LabQuest 2, see LabQuest 2 FAQs
There have been reports of LabQuest units freezing when the units are in a low battery condition and then connected to AC power. Symptoms of the freeze are:
* Screen is on but not responsive to taps (may show battery warning message)
* Blue power light is on but power button will not work.
* Hard keys are not responsive.
The fix is to do a hard reset. See TIL 1777: How do I reboot or reset a LabQuest?