The following OHAUS balances are discontinued but may still be supported by some Vernier software (see Which OHAUS balances can I use with Vernier technology?):
Scout Pro 0.01 g Balance (200 g) (OHSP-202, discontinued)
Scout Pro 0.01 g Balance (400 g) (OHSP-402, discontinued)
Scout Pro 0.001 g Balance (120 g) (OHSP-123, discontinued)
Ohaus Adventurer Pro 0.001 g Balance (300 g) (OHAP-313, discontinued)
Ohaus Adventurer Pro 0.0001 g Balance (110 g) (OHAP-114, discontinued)
For OHAUS Scout balances, use OHAUS Scout® USB Cable (OHS-USB)
For OHAUS Scout Pro balances, use Scout Pro USB Connection Kit (OHSP-USB)
-Primary Test: Plug in the balance and run the data-collection program. The software should automatically identify the balance and start reading live mass values.
-Secondary Test: Reset the data collection software by either starting a New File or quitting and restarting the program.
Frequently Ask Questions (FAQs)
How do I use an OHAUS balance with Logger Pro software?
How do I use an OHAUS balance with a LabQuest?
OHAUS Balance Troubleshooting
Which OHAUS balances can I use with Vernier technology?
Logger Pro 3 can't find my OHAUS balance.
Why is the sampling rate for Ohaus balances limited to 1 sample/sec?
Do OHAUS balances and Watts Up devices work with Logger Pro 3 and 64 bit versions of windows?
My OHAUS balance is inaccurate. How do I fix it?
Why doesn't the number of decimal places on my OHAUS balance match the number of decimals displayed in the software?