National Instruments has posted an example LabVIEW VI for using a Vernier temperature sensor with NI ELVIS. The example is currently posted in the NI Developer Zone with an article titled “Interfacing VIs for Vernier Biosensors”. If you run the example with our temperature sensor connected via the BTA-ELV adapter the readings will be off. The problem is that the VI converts from raw volts to temperature units using a resistor value of 20k. The BTA-ELV actually has a resistor with a value of 15k. Making this change in the VI will fix the problem.