A brand new O2 Gas Sensor does not need to be calibrated, but the sensor will drift over time and will need to be calibrated.
To calibrate the sensor, follow the normal 2-point calibration procedure.
-For the first point, push and hold the calibrate button on the sensor using a paper clip or LabQuest stylus.
-Enter a value of 0 for this reading.
-Release the button and take a second reading in air.
-Enter a value of 20.9% oxygen or a corrected value from the table in the sensor booklet.
Since the timing of when you press and release the calibrate button on the sensor is important, the video below should be every helpful:
TIL 1421: O2 Gas Sensor Troubleshooting and FAQs
TIL 1605: Can my O2 Gas Sensor (O2-BTA) be repaired?