Hand Dynamometer (HD-BTA)
-Primary Test: Confirm that the sensor was properly zeroed. If the sensor does not read zero in the orientation in which you are using it, follow the standard zeroing procedure for the data-collection program you are using.
-Secondary Test: Confirm that you are using Graphical Analysis, Logger Pro 3.3 or newer or Logger Lite 1.3 or newer.
RELATED TIL ENTRIES
-Accuracy: ±0.6 N
-Power: 7 mA @ 5VDC
-Typical Resolution: 0.2141 N
-Safety range (maximum force without damage done to sensor): 0 to 900 N
-Operational range: 0 to 600 N
-Calibrate? No. The sensor is custom calibrated before shipping.
-No calibration process is given in the sensor booklet.
Default calibration formulas are below:
Default calibration (N)
Default calibration (kg)
Default calibration (lbs)
RELATED VERNIER PRODUCTS