Chloride Ion-Selective Electrode, order code CL-BTA, $199
Specifications and User Guide
-Primary Test: Check the raw voltages while calibrating in the standards. The high (1000 mg/L) should be around 2.0 volts. The low (10 mg/L) should be around 2.8 volts.
-Secondary Test: Refer to TIL 665: General tips for using Ion Selective Electrodes (ISE).
RELATED TIL ENTRIES
-TIL 1060: Do any of your sensors require special storage?
-TIL 759: Do you sell ISE Standards?
-TIL 665: General tips for using Ion Selective Electrodes (ISE).
-TIL 1296: How can I have my ISE read mV output instead of mg/L?
-TIL 1399: Can I use Vernier water quality sensors in salt water or in a salt water aquarium?
-TIL 3389: How do I sterilize an Ion-Selective Electrode?
RELATED VERNIER PRODUCTS
-Standard Low Cl ISE Solution, order code CL-LST, $18
-Standard High Cl ISE Solution, order code CL-HST, $18
Fill out our online support form or call us toll-free at 1-888-837-6437.