When collecting blood pressure data using a Blood Pressure Sensor (BPS-BTA), and LabQuest 2 running LabQuest App 2.8.1, the calculated values for Systolic, Diastolic, Pulse, and Mean Arterial Pressures remain constant after the first run. The graph continues to change with each new run, as does the value for Cuff Pressure, but the other calculated values do not appear to update.
The work around for this issue is to be sure to store each run of data. Once you have collected additional runs of data, you can tap on Run and select the run that you would like to display. This will update the meter display so that the calculated values for the run you have selected are visible.