-Primary Test: Check the raw voltages while calibrating in the standards. The high (100 mg/L) should be around 2.1 volts. The low (1 mg/L) should be around 1.3 volts.
-Secondary Test: Refer to General tips for using Ion Selective Electrodes (ISE).
RELATED TIL ENTRIES
–Do any of your sensors require special storage?
–Do you sell ISE Standards?
–General tips for using Ion Selective Electrodes (ISE).
–How can I have my ISE read mV output instead of mg/L?
–What Is the Difference Between Auto-ID Sensors, Non-Auto-ID Sensors, and Smart Sensors?
–How do I sterilize an Ion-Selective Electrode?
-Order code: NH4-BTA
-Range: 1 to 18,000 mg/L or ppm
-pH range: 2 to 7 (no pH compensation)
-Interfering ions: K+, Li+, Na+, Cs+, Mg3+, Ca2+, Sr2+, Ba2+
-Electrode slope: +55 +/- 3 mV/decade
-Approximate calibration voltages: High (100 mg/L) 2.1 V; Low (1 mg/L) 1.3 V
-Electrode resistance: 0.1 to 5 MΩ
-Accuracy: ± 10% of full scale (calibrated 1 to 100 mg/L)
-Temperature range (can be placed in): 0 to 40°C (no temperature compensation)
-Minimum immersion: 1.1 inch
-Electrode length: 155 mm
-Body diameter: 12 mm
-Cap diameter: 16 mm
-Cable length: 100 cm
-Calibrate? Yes. Conduct a two-point calibration, using the standard solutions that are shipped with the sensor. First point: The ISE should be soaking in the High Standard. Enter the concentration value of the High Standard. Second point: Remove the ISE from the High Standard, rinse well with distilled water from a wash bottle, and gently blot dry with a paper towel or lab wipe. Place the electrode into the Low Standard. Enter the concentration value of the Low Standard.
-Important: Make sure the ISE is not resting on the bottom of the container and that the small white reference contacts are immersed. Make sure no air bubbles are trapped below the ISE. Hold the ISE still and wait for the voltage reading to stabilize. Stirring the solution during this process is ideal.
RELATED VERNIER PRODUCTS
-ISE Ammonium Replacement Module, Ammonium Replacement Module (NH4-MOD)
-ISE Ammonium Low Standard, Standard NH4 ISE Solution (NH4-LST)
-ISE Ammonium High Standard, Standard NH4 ISE Solution (NH4-HST)