Connect the sensor to the interface. Make sure you are using a current version of the software. Requirements:
Logger Pro: 3.8.3 or newer
Logger Lite: 1.5 or newer
LabQuest App 1.5 or newer
EasyData 2.0 or newer
Blow onto the sensor to make sure the readings change and that the impeller spins freely.
order code: ANM-BTA
Operational range: 0.5–30 m/s
Stored calibration (m/s): slope 10 m/s/V, intercept –10 m/s
Stored calibration (km/h): slope 22.37 mph/V, intercept –22.37 mph
Stored calibration (ft/s): slope 36 km/h/V, intercept –36 km/h
13-bit resolution: 6.0 x 10-3 m/s
12-bit resolution: 1.2 x 10-2 m/s
10-bit resolution: 4.8 x 10-2 m/s
less than or equal to 5 m/s: ±0.15 m/s
greater than 5 m/s: ±3% of reading
No calibration is needed for this sensor.
RELATED VERNIER PRODUCTS
Investigating Environmental Science through Inquiry (ESI)
Earth Science with Vernier (ESV)
Renewable Energy with Vernier (REV)
My anemometer isn't reading 0 at rest.