– Primary Test: It should read approximately 4 in its storage solution.
– Secondary Test: Take readings in fresh buffers, NOT distilled water! The pH reading should be within 0.2 pH units of the pH buffer. For example, if using a pH 4.0 buffer, your reading should be between 3.8 and 4.2.
If no buffers are available, try it in vinegar (~ pH 2.5-3.5) and ammonia (~ pH 10.5-11.5). If readings do not change when the sensor is placed in different solutions, the sensor is possibly defective or damaged.
RELATED TIL ENTRIES
–Which pH Sensor Should I Buy?
–Why won't my pH Sensor calibrate?
–Where do I get pH buffer solutions?
–What is the difference between the pH buffer capsules with order codes PHB and PH-BUFCAP?
–How do I change the displayed measurement channel on my Go Direct Electrode Amplifier or ISE Amplifier?
–My Go Direct pH arrived in two pieces, how do I put it together?
–The glass bulb of my pH Sensor is broken. Can it be repaired?
–The pH electrode I have been using no longer seems to work or is slow to respond. Is there anything I can do to fix it?
–My pH probe no longer gives an accurate reading. What should I do?
–Why doesn't my pH Sensor read pH 7 in distilled or deionized water?
–What type of liquids cause problems for the pH Sensor?
MAINTENANCE AND STORAGE
–Where do I get pH or ORP Storage Solution? (Buffer Storage Solution)
–What do I do about mold growth in my bottle of pH electrode storage solution?
–How can I keep my pH or ORP storage solution from spilling when the sensor is in use?
–My pH sensors were stored dry. Will they still work?
–What is the best way to clean the glass bulb of the pH electrode?
–Do any of your sensors require special storage?
–I bought your pH buffer capsules. Can I make storage solution for my pH Sensors with this kit?
–How do I dispose of my pH sensor?