Go Direct® Hand Dynamometer (GDX-HD)
-Primary Test: Confirm that the sensor was properly zeroed. If the sensor does not read zero in the orientation in which you are using it, follow the standard zeroing procedure for the data-collection program you are using.
-Secondary Test: Confirm that you are using Graphical Analysis 4.7 or newer.
RELATED TIL ENTRIES
Hand Dynamometer Troubleshooting and FAQs
-Calibrate? No. The sensor is set to the stored calibration before shipping.
-No calibration process is given in the sensor booklet.
If the sensor is not reading zero in the orientation in which you are using it, follow the procedure below:
1. Hold the Hand Dynamometer in the same orientation you plan to use it. Do not put any force on the pads of the
2. When the readings stabilize, click or tap the Force meter and choose Zero. The readings for the sensor should be close to zero.
RELATED VERNIER PRODUCTS
Hand Dynamometer (HD-BTA)