Stainless Steel Temperature Probe, order code TMP-BTA, $29
Specifications and User Guide
-Primary Test: Plug in the sensor and run the data-collection program. Hold the tip of the sensor in your hand and check for the temperature readings to change.
-Secondary Test (calculator only). If the main screen of DataMate displays -999.9 as the temperature reading, press the CLEAR key to reset the program.
The most common causes for this sensor to read incorrectly are:
1- Twisting the cable: Sometimes students twist or crimp the wire near the handle of the sensor. Over time, this can cause the wires to come loose and make the sensor stop working.
2- Overheating: Measuring the temperature of a flame or the surface of a hot plate will damage the sensor.
3- Melted cable: Draping a cable across a hot plate will melt the covering and damage the sensor.
4- The unit is not waterproof: Water can seep into the hilt of the sensor and damage the electronics. Only submerge the stainless steel portion of the sensor.
Unfortunately, these causes of breakage are considered misuse and are not covered by our warranty. Repair is not possible for this type of damage.
RELATED TIL ENTRIES
-TIL 2345: Why does my temperature sensor read a value that does not change and is very high/low?
-TIL 3805: My temperature sensor reads 80C at room temperature on a LabQuest 2.
-TIL 2285: The tab on the BTA or BTD connector of my sensor is broken. Can it be repaired?
-TIL 2377: How should I wrap my sensor cable?
-TIL 1310: Can you calibrate a Stainless Steel Temperature Probe?
-TIL 1399: Can I use Vernier water quality sensors in salt water or in a salt water aquarium?
-Range: -40 deg C to 135 deg C
-Maximum temperature that the sensor can tolerate without damage: 150 deg C
-Typical accuracy: +/- 0.2 deg C at 0 deg C, +/- 0.5 deg C at 100 deg C
-12-bit resolution: (with LabPro or Go! Link) 0.17 deg C (-40 to 0 deg C); 0.03 deg C (0 to 40 deg C); 0.1 deg C (40 to 100 deg C); 0.25 deg C (100 to 135 deg C)
-10-bit resolution: (with CBL or CBL 2) 0.68 deg C (-40 to 0 deg C); 0.12 deg C (0 to 40 deg C); 0.4 deg C (40 to 100 deg C); 1.0 deg C (100 to 125 deg C)
-Temperature sensor: 20 kΩ NTC Thermistor
-Response time: (time for a 90% change in reading)
10 seconds in water, with stirring
400 seconds in still air
90 seconds in moving air
Length (handle plus body): 15.5 cm
Stainless steel body: length 10.5 cm, diameter 4.0 mm
Probe handle: length 5.0 cm, diameter 1.25 cm
Does not work with ULI and SBI
-If you wish to calibrate this sensor, detailed instructions can be found in TIL 1310: Can you calibrate a Stainless Steel Temperature Probe?
RELATED VERNIER PRODUCTS
-Extra Long Temperature Probe, order code TPL-BTA, $99
-Wide-Range Temperature Probe, order code WRT-BTA, $82, (for use with temperature range of –20 to 330 deg C (–4 to 626 deg F)
-Thermocouple, order code TCA-BTA, $69, (for use with super-cooled liquids and flame)
-Surface Temperature Sensor, order code STS-BTA, $23, (quick response - for use in air only)
-Infrared Thermometer, order code IRT-BTA, $199, (non-contact temperature meaurements)
-Go!Temp, order code GO-TEMP, $39, (direct USB connection to computers or LabQuest)