-Primary Test: Connect an ORP sensor or a pH sensor to the Electrode Amp and connect the Electrode Amp to a Vernier interface. Start the data-collection program. Place the ORP or pH sensor in a buffer solution of known pH and confirm that the readings are within range.
-Secondary Test: Confirm that the BNC connection is secure.
-Calibrate? Normally, yes. However, it depends on the sensor that is connected. The Electrode Amplifier amplifies the voltage produced by an electrode into a range where it can be monitored by the lab interface.
The Electrode Amplifier will report values in units of millivolts. If a pH electrode is attached, however, you can choose to read in units of pH. Custom, linear 2-point calibrations can also be performed if desired.
-You can open an experiment file in LoggerPro which will use the stored calibration in the software.