Thermocouple, order code TCA-BTA, $69
Specifications and User Guide
-Primary Test: Connect the Thermocouple to an interface and check the accuracy of the readings for temperature in room air. They should be within 3 deg C.
-Secondary Test: Place your fingers over the wires and ensure that the temperature responds.
RELATED TIL ENTRIES
-TIL 1529: Can I purchase a replacement thermocouple wire for my TCA-BTA sensor?
-TIL 1303: Can I use our thermocouple in a bunsen burner flame?
-TIL 1292: Why doesn't my new Thermocouple work with Logger Pro or Logger Lite?
-Order code: TCA-BTA
-Typical accuracy: 0 to 900 degC: +/- 3 degC; -200 to 0 degC: +/- 5 degC; 900 to 1400 degC: +/- 14 degC
-Resolution, 12-bit: 0.40 degC
-Power: 11 mA @ 5 VDC
-This sensor is set to the stored calibration before shipping.
-If you wish to calibrate the sensor, you must conduct a three-point calibration using Logger Pro. If you choose to store this calibration to the sensor, you will have to overwrite either the high or the low factory-installed calibrations.
To access these different ranges in Logger Pro or Logger Lite:
1. From the Experiment menu, select Set Up Sensors.
2. Select your interface. This will open a new dialog box.
3. Click on the sensor icon button to display the drop down menu and select the temperature setting you would like to use.
Other platforms (LabQuest or Graphical Analysis) do not currently allow you to switch between settings.
Units shipped before September 2016 have the following calibration equations (see TIL 1292: Why doesn't my new Thermocouple work with Logger Pro or Logger Lite? for more information):
Thermocouple degC Low
Thermocouple degC High
RELATED VERNIER PRODUCTS
-Stainless Steel Temperature Probe, order code TMP-BTA, $29
-Extra Long Temperature Probe, order code TPL-BTA, $99
-Wide-Range Temperature Probe, order code WRT-BTA, $82, (for use with temperature range of -20 to 330 deg C (-4 to 626 deg F))
-Surface Temperature Sensor, order code STS-BTA, $23, (quick response - for use in air only)
-Infrared Thermometer, order code IRT-BTA, $199, (non-contact temperature measurements)
-Go!Temp, order code GO-TEMP, $39, (direct USB connection to computers or LabQuest)