Wireless Dynamics Sensor System (WDSS, discontinued)
Specifications and User Guide
- Primary Test: There is extensive information on installing Bluetooth radios and establishing radio communications with WDSS, see
Bluetooth Help and Troubleshooting Tips related to the Wireless Dynamics Sensor System (WDSS)
- Secondary Test: Once communication is established, do the following:
- Choose Set Up Sensors –> WDSS:(name) from the Experiment menu.
- Click Identify; Run light will blink on WDSS.
- Click Manage Memory; you will see any stored runs existing on the device.
Check the Delete boxes and click OK if you want to clear the memory; otherwise just click OK.
- Enable all five sensors by placing a check in each checkbox, then click Close. You should see five meters on screen.
- Push and pull force sensor; does reading change?
- Hold WDSS so each of x, y and z axes are up; does meter read about 9.8 m/s2 in each case?
- Altimeter will read near zero, but cannot be easily tested indoors.
Note that if you last used a particular WDSS with, only some of the channels active, the next time that computer detects that WDSS, those channels will be active.
RELATED TIL ENTRIES
Logger Pro sees my WDSS by name, but then fails to collect data.
Bluetooth Wireless FAQs
How good is the time sync between WDSS and other interfaces?
Will LabQuest work with the Wireless Dynamics Sensor System?
Why is the WDSS run LED red?
Is there a way to force my bluetooth radio to load the Microsoft driver?
How good is synchronization between two WDSS? Good enough for impulse experiments?
Where is the accelerometer located inside the WDSS?
Can WDSS communicate with Graphical Analysis 4 or with an iOS device?
Problems with force sensor giving bad readings because hex barrel has been removed and improperly reinstalled.
What should I purchase for doing Amusement Park Physics?
How do I connect a Wireless Dynamics Sensor System (WDSS) to a PASCO cart?
Can I use an accelerometer to measure velocity or position?
My force sensor fell into the floor, and now the hex bolt sheared off inside the load cell.
Dimensions: 121 mm x 53 mm x 39 mm
Mass: about 200 g, depending on battery type used and attachments
Batteries: 3.7 volts, 1600-1800 mAh, Li-Ion rechargeable with safety circuit
Lithium ion batteries self-discharge about 5 to 10% of its stored charge each month until approximately half depleted. This is normal. Disconnecting the battery will not change this.
Operating Current: 35-50 mA @ DC minimum 3.2V
Nominal Charging Current: @350 mA
DC input, Rating: DC 6V 500mA maximum (Max. with battery charging and unit operating)
Bluetooth radio frequency: approximately 2.45 GHz (2450 MHz), with frequency hopping, so the frequency is not fixed
Wireless Range: Up to 10 meters (30 ft) with no obstruction.
Number of WDSS devices that can be connected to one computer: 7
- Force Sensor
Range: ±50 N
Frequency Response: 0-100 Hz
Resolution: 0.006 N(<10 N), 0.03 N(>10 N)
- Accelerometers (for each of the three accelerometers):
Range: -60 to +60 m/s^2
Accuracy: ±0.4 m/s^2 (±0.05 g)
Frequency Response (maximum): 100 Hz
Temperature Range: -40 to +85°C
Accelerometer location: 54 mm from the force sensor end of the base, 24 mm above the bottom, and on the midline.
Altitude change range: ±200 m (1000 ft)
Sensitivity: 1 meter
Combined linearity and hysteresis: typical ±0.1% full scale, maximum ±0.5% full scale
Response time: 0.5 second
Calibrate this sensor? No, calibration of this sensor is not normally needed or advised. The sensor is set to the stored calibration before shipping. See Specifications and User Guide for detailed calibration information.