– Primary Test: Press the power button on the sensor to turn it on. (The red LED should flash.) Start the Graphical Analysis app on your device and select the sensor from the list of available sensors. (The green LED will start flashing.) Does the sensor read near zero when no force is applied? Does the sensor read a positive value when pulling, and a negative value when pushing?
-Force Sensor Test: Attach an eye bolt to the force sensor. Lift the entire Go Direct Structures & Materials Tester by the eye bolt. The force sensor should read approximately 65 – 70 N.
Displacement Sensor Test: Note the location of triangle mark on the wheel. Zero the displacement sensor. Turn the wheel one complete rotation. The displacement should read 0.159 cm. Rotate the wheel back to its original position. Verify that it reads 0.00 cm.
Is the tackle kit for Vernier Structures & Materials Tester (VSMT-TK) able to handle 1000 N of force?
Why is my VSMT force sensor stuck at a high reading (>1000 N)?
Trust Tester Accessory for Structures & Materials Tester Troubleshooting and FAQs
Can I use my Structures & Materials Tester to conduct 3-point and 4-point flexural testing?
Can I use the Structures Tester for the Science Olympiad Boomilever event?
How do I get the Beams to Bridges e-book?
How can I use my Structures & Materials Tester with Graphical Analysis Pro?
I'm missing parts to fasten the endcaps onto my Go Direct Structures & Materials Tester. Can you send me replacements?
See user manual.
The VSMT Force sensor should not require calibration. A 2-point calibration is recommended if necessary.
The VSMT Displacement sensor cannot be calibrated.
RELATED VERNIER PRODUCTS
VSMT Tackle Kit (VSMT-TK)