The Differential Voltage Probe is designed for exploring the basic principles of electricity. Use this probe to measure voltages in low voltage AC and DC circuits. With a range ±6.0 V, this system is ideal for use in “battery and bulb” circuits.
Note: Vernier products are designed for educational use. Our products are not designed nor are they recommended for any industrial, medical, or commercial process such as life support, patient diagnosis, control of a manufacturing process, or industrial testing of any kind.
Compatible Software and Interfaces
Choose a platform below to see its compatibility requirements.
The Differential Voltage Probe is fully supported with LabQuest 2 (version 2.8.5), a standalone data logger with built-in graphing and analysis software.
CBL 2 and LabPro interfaces cannot be used with TI-84 Plus CE calculators.
DataMate cannot be used with color screen TI-84 Plus calculators; use EasyData with these calculators.
Connect the sensor to the interface (LabQuest Mini, LabQuest 2, etc.).
Start the appropriate data-collection software (Logger Pro, Logger Lite, LabQuest App) if not already running, and choose New from File menu.
The software will identify the sensor and load a default data-collection setup. You are now ready to collect data.
If you are collecting data using a Chromebook™, mobile device such as iPad® or Android™ tablet, or a Vernier wireless sensor or interface, please see the following link for up-to-date connection information:
Connect the sensor following the steps in the Getting Started section of this user manual.
The Differential Voltage Probe is designed to be used like a voltmeter. The leads should be placed across a circuit element. The differential input range is –6 volts to +6 volts. Over‑voltage protection is provided so that slightly higher voltages will not damage the sensor. You should NEVER use high voltages or household AC with this probe.
You should not have to perform a new calibration when using the Differential Voltage Probe in the classroom. We have set the sensor to match our stored calibration before shipping it. You can simply use the appropriate calibration file that is stored in the data-collection program from Vernier.
If you choose to calibrate the Differential Voltage Probe for increased accuracy, use the standard, two-point calibration procedure. Another option to consider instead of calibrating is “zeroing” the sensor. This is done by shorting out the leads of the sensor, then choosing the Zero option in the data-collection software. This option adjusts the calibration offset but does not adjust the calibration gain.
Differential Voltage Probe input voltage range
Maximum voltage on any input
Input impedance (to ground)
Supply current (typical)
Output voltage range
Vo = –0.4 (V+ –V–) + 2.5
Default calibration values
slope: –2.5 V/V
intercept: 6.25 V
How the Sensor Works
The Differential Voltage Probe measures the potential difference between the V+ clip (red) and the V– clip (black). The voltage probes have differential inputs. The voltage measured is with respect to the black clip and not circuit ground. This allows you to measure directly across circuit elements without the constraints of common grounding. The voltage probes can be used to measure negative potentials, as well as positive potentials.
The output of this system is linear with respect to the measurement it is making. A built-in amplifier allows you to measure positive and negative voltages on any of our interfaces. Since many lab interfaces can read voltages only in the range of 0 to 5 volts, the amplifier offsets and amplifies the incoming signal so that the output is always in the range of 0 to 5 volts. If an input is zero volts, for example, the amplifier will produce an output of 2.5 volts. The output varies from this 2.5 volt level, depending on the input.
If the Differential Voltage Probe is not operating as expected, check the following:
Are the leads properly and securely connected? The probe is designed to be used like voltmeter leads. It should be placed across a circuit element.
Is the sensor voltage fluctuating? Try a DC power source, such as a battery. When measuring DC voltages with power supplies, some power supplies do not provide a steady DC signal. If the sensor reading is correct when using a battery, the problem may be the power supply.
If you have watched the related product video(s), followed the troubleshooting steps, and are still having trouble with your Differential Voltage Probe, contact Vernier Technical Support at email@example.com or call 888-837-6437. Support specialists will work with you to determine if the unit needs to be sent in for repair. At that time, a Return Merchandise Authorization number will be issued and instructions will be communicated on how to return the unit for repair.
Vernier warrants this product to be free from defects in materials and workmanship for a period of five years from the date of shipment to the customer. This warranty does not cover damage to the product caused by abuse or improper use. This warranty covers educational institutions only.
Privacy & Cookies Policy
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.